The IEC 62052-11 has a tiny little detail about the power consumption in the current circuit.
Power consumption is part of the technical losses. The utilities are bearing these costs. The end-user is not charged.
A test bench consists of several elements which can introduce additional variable errors.
An energy meter has certain specifications, including maximum current. This is the base for all kind of meter type tests.
The IEC 62052-11 Edition 2 has introduced some changes for the meter marking and documentation. Today we cover the mandatory content of the nameplate for IEC meters.
The IEC 62052-11 Ed.2 comes with a lot of changes for meter type testing. Another new introduced test for electromagnetic compatibility (EMC) is the ring wave immunity test.
Sometimes it happens that you touch the door knob and you get an electrical shock. It’s not dramatically for your body but an unpleasant feeling.
Meter manufacturers, utility laboratories and type testing institutes have planning security until year 2025.
The critical change value indicates the maximum amount of change allowed in the meter’s energy registers during disturbance tests without any current flowing in the meter’s current circuits.
Each electronic energy meter must have at least one optical test output. Commonly its named with flashing pulse diode or similar.