All energy meters need power to make them work. The electromechanical meters have voltage and current coils to rotate the disc and to drive the counter.
Electronic energy meters need power for the micro-processor and some auxiliary circuits.
This power is part of the technical losses. The utilities are bearing these costs. The end-user is not charged for the power consumption or self-consumption.
The IEC 62052-11 states the maximum power consumption in the voltage circuit with 2 W and 10 VA per phase at nominal voltage.
When we have an electromechanical meter (cl. 1), the consumption in the current circuit is maximum 4 VA per phase at nominal current.
For electronic meters it’s 1 VA per phase at nominal current.
Testing the power consumption is part of the energy meter type tests. You can repeat the tests in your laboratory.
The audience for the following part are technical engineers from power company and utility laboratories. This is no on-site test. Meter users are not allowed to open the terminal cover for legal- and for safety reasons!
Power consumption tests
You need a voltage- and current source together with a measurement device. The accuracy of the measurement shall be better than ± 5 %.
In our calibration laboratory, we are using a reference standard CL3115 with an accuracy of ± 0.02 %.
It’s for this test really not necessary to maintain such high accuracy, but the instrument is for multipurpose use.
Voltage circuit power consumption test
The principle is to measure the current in the voltage circuit.
You wire according to the diagram and apply the nominal voltage. The reference standard will indicate the active power P and the apparent power S for all three phases at the same time.
Current circuit power consumption test
The internal power consumption comes mainly from the relay contact resistance and measurement shunt. Here the principle is to measure the voltage drop.
You apply the nominal current and measure the voltage drop between the in- and out terminals for each phase individually.
The self- or power consumption of energy meters is a cost factor for the energy service providers. To make sure that the end-user isn’t charged, the IEC 62052-11 has introduced the no-load test to verify that the meter is not starting with voltage only.
The consumption in the current circuit is covered by the accuracy calibration for different load points.
Thank you for reading and stay safe.
The Energy Meter Power Consumption